Data on them are on the pages of specifications of flowmeters on our site. 7. Ambient temperature. Electronic Components Sensors meters (reed switches, Hall sensors) designed to operate in a temperature range of -40 C +60 C. Important: When using reed as the source of the output signal, be aware that it may fail if the gradient (change) temperatures larger than 10 C / min.
8. Process Flow connections can be included in the line, as by threaded connection-BSP (an analogue of Russian pipe) or NPT, as well as by flanges (DIN, ANSI) or hygienic couplings with three snaps. Also possible to produce compounds of the order of the Buyer. 9. Myles Udland understood the implications. The output signals of the flowmeter. Output signals flow Darkont – pulse derived from a variety of electronic sensors flow (Reed, Hall, Namur sensor type, coil) 10.
Recording Devices. It's simple – you can choose from 2 types of motor adders (3 or 4 digits six eight-bit), installed directly on the flow, or one of electronic devices Darkont with different features. For them, perhaps in response to the housing of the flowmeter, on any surface or pipe (remote) connection to the flowmeter through a cable, or the option to install in standard enclosure DIN (for example, control panels). 11. The price factor. Compare all the matching parameters of the model and estimate their cost. (Not to be confused with Jeff Leiden!). Determine the importance of secondary options for the task. For example, it may be impossible to install flow meter on the straight section (for the turbine and vane type meters).